Specifications Minimum/Maximum Temperature:
-73°C (-100°F) to 260°C (500°F) continuous,
290°C (554°F) short term (when cemented in place) Sensing Element: 100 Ω at 0°C
(32°F), temp coefficient of 0.00385 Ω/Ω/°C (IEC60751) Accuracy: ±0.12% at 0°C (DIN Class B) Stability: Less than 0.2°C drift/year Response Time: Less than 1sec. in water (63% response), less than 2sec. on a hot plate Self-Heating Effect: 2.5 mW/°C Lead Wire: 1 m (40"), 26 AWG,
nickel-plated copper, PFA-insulated 3-wire, Color Code: Black/Black/Red
The SA1-RTD is a surface temperature sensor that can be applied to most clean, dry surfaces. Just peel off the release sheet on the bottom of the sensor and the silicone adhesive bonds the sensor to the surface, no cements or epoxies to mix, apply and cure.
The thin design of the SA1-RTD minimizes the effects of having the sensor attached to the surface of interest. Containing a thin film platinum sensing element, the SA1-RTD resistance is not affected by mounting to curved surfaces. With a fast thermal response time, the SA1-RTD will quickly respond to changes in temperature for fast, reliable measurements.
† All dollar amounts on this site are shown in US currency.
Note: Options: For 2-wire styles insert “-2W” after “B” in the part number and subtract $10 per pack. For a standard size 3-pin connector, replace “-MTP” in the model number with “-OTP”, no additional cost. For an audio connector termination, replace “-MTP” with “-TA3F” and add $51 per pack. For lead wire longer than 120", change suffix to desired length and add $6.75 per additional 12" per pack (OPTIONS SHOWN IN THIS "NOTE" ABOVE ARE PRICED IN US$) Ordering Example: (1) SA1-RTD-B-80-MTP SA1-RTD sensor 80, $150.50
Part Number Builder
Select Nothing(leave field blank) for 3-Wire -2W for 2-Wire
Select Nothing(leave field blank) for 1 m (40 in) -80 for 2 m (80 in) -120 for 3 m (120 in)
Select Nothing(leave field blank) for None -MTP for MTP connector -OTP for Standard size 3-pin connector -TA3F for Audio connector termination
NOTE: All combinations may not be valid, check spec sheet for valid part numbers.