The CT87P pressure recorder is a versatile instrument that accurately measures and records pressures. The remote sensor has a 6' long cord and a 17-4 stainless steel pressure port with ¼NPT threads. Burst pressure 2500 psi rated for any gas or fluid compatible with 17-4 stainless steel. Battery backup assures normal operation during power interruptions. The unit can also be run from any 12 Vdc source (auto or marine).
SPECIFICATIONS Pressure Ranges psi Model: 0 to 500 psi, 0 to 150 psi, 0 to 50 psi Pressure Ranges bar Model: 0 to 35 bar, 0 to 10 bar, 0 to 3.5 bar Pressure Accuracy: ±1% FS Ambient Operating Temp Range: 0 to 60°C (32 to 140°F) Remote Pressure Sensor: 1.8 m (6') long 17-4 stainless steel ¼ NPT pressure port, 2500 psi burst pressure, rated for any gas or fluid compatible with 17-4 stainless steel Ambient Operating Relative Humidity: 96% maximum Selectable Chart Speeds: 6 hours, 24 hours, 7 days, 31 days Chart Rotation Mode (Selectable): Single turn or continuous Chart Speed Accuracy: ±1% Chart Diameter: 152 mm (6') Display: 3-digit, 13 mm (0.5') LED Primary Power Supply: 120 Vac, 50/60 Hz, or 220 to 240 Vac, 50/60 Hz (AC adaptor included) Battery Life: 48 hours continuous use Battery Backup: 8 'AA' alkaline batteries (included) Recording Pen: Ink Case Size: 235 H x 184 W x 70 mm D (9¼ x 7¼ x 2¾') Weight: 2.13 kg (4.7 lb) Case Material: Polycarbonate
Rated 1 out of
Dave 53 from
Chart recorderI bought this about a month ago. It's a nice unit, but the chart skips. I don't get a continuous mark for 24 hours.
Date published: 2018-06-18
Rated 5 out of
works wellHave one in continuous operation for over 14 years...still running great. Easy to use and setup. Appropriate for an indoor installation.
Date published: 2019-03-28
Rated 1 out of
Never use them!We bought four of them. None of them is working satisfactory. Pen is vibrating and very thick line is generated. Actual reading is difficult to find out.
It uses black pen and chart lines are dark brown, so you can not visualize properly actual reading line.
It stops recording in between and again start during week cycle.
I would never use it again.