NOT ALL MODELS ARE STANDARD WITH VALVES, VERIFY MODEL DESCRIPTION TO CONFIRM
The FL-2000 Series offers a wide variety of precision flow meters for use in medical, industrial, chemical, and laboratory applications at an economical price. Units are available with or without valves.
SPECIFICATIONS Accuracy: Models FL-2001–FL-2025: ±5% F.S. Models FL-2031–FL-2069: ±3% F.S. Models FL-2071–FL-2128: ±2% F.S. Float: Black glass stainless steel Body: Clear acrylic Seals: Buna O-Rings with brass or PVC fittings FKM O-Rings with stainless steel fittings Pressure: 100 psig max @ 21°C (70°F) Temperature: 65°C (150°F) max @ 0 psig Fittings: Brass std; stainless steel optional except for FL-2071 through FL-2128, which have 1 NPT PVC fittings only Valves: Models FL-2001 through FL-2069: brass standard; stainless steel cartridge type (optional) FL-2071 through FL-2128: optional plastic in-line gate
What is the pressure drop of a FL-2012 (0.2 to 2.5) flow meter?
Asked by: jenn
Thank you for your question?
The max drop is 3 PSI for the FL-2012
Answered by: application engineer je
Date published: 2019-07-01
Can I use an equation to determine the flow at non-standard pressure and temperature? On another MFG's site I saw Q2 = Q1 * sqrt (P1*T2/(P2*T1)). Is that equation correct for this flowmeter?
Asked by: Engineer2020
Send us an email to firstname.lastname@example.org. We can provide correct equation on a PDF form.
Answered by: ASE HPO
Date published: 2021-12-11
Can these be used with inert gases or solely just air?
Asked by: amyfng
Hi, thanks for connecting with us! You can use these meters with inert dry gases other than air but you will have to correct for the specific gravity. If you need help calculating this, please contact our Flow Department at email@example.com.
Answered by: Jane
Date published: 2019-05-20
We are using FL-2012 not with a ball float but with a conical float. Do we have to determine the flowrate by reading the center, top or bottom of the conical float?
Asked by: bob100
Thank you for your inquiry! For conical floats the reading will be taken from the top