The FMA-1600A Series mass and volumetric Omega's flow meters use the principle of differential pressure within a laminar flow field to determine the mass flow rate. A laminar flow element (LFD) insdie the meter forces the gas into laminar (streamlined) flow. Inside this region, the Poiseuille equation dictates that the volumetric flow rate be linearly related to the pressure drop along a fixed distance of the LFE. This, along with the viscosity of the gas, is used to accurately determine the volumetri flow rate. Separate absolute temperature and pressure sensors are incorporated and correct the volumetric flow rate to a set of standard conditions. This standardized flow rate is commonly called the mass flow rate and is reported in units such as standard cubic feet per minute (SCFM) or standard liters per minute (SLM).
Standard units include a 0 to 5 V output (4 to 20 mA optional) and RS232 communications. The gas select feature can be adjusted from the front of the keypad or via RS232 communications. Volumetric flow, mass flow, or absolute pressure, and temperature can be viewed or recorded through the RS232 connection. It is also possible to multi-drop up to 26 units on the same serial connection to a distance of 125'. These flowmeters are also available in a portable version for use in the field (-B models).
SPECIFICATIONS Accuracy: ±(0.8% of rdg + 0.2% FS) Repeatability: ±0.2% Turndown Ratio: 200:1 Response Time: 10 ms typical default response time for 63.2% of a step change. A variable register allows response time to be field adjustable to a certain extent via RS232 communications. The primary trade-off for response time is signal noise Output: 0 to 5 Vdc standard Operating Temperature: -10 to 50°C (14 to 122°F) Zero Shift: 0.02% FS/°C/atm Span Shift: 0.02% FS/°C/atm Humidity Range: 0 to 100% non-condensing Pressure (Max): FMA-1600A Series: 145 psig FMA-1600A Series: 50 psi differential FVL-1600A Series: 10 psig FVL-1600A Series: 10 psi differential Measurable Flow Rate: 125% FS Supply Voltage: 7 to 30 Vdc (15 to 30 Vdc for 4 to 20 mA output) Supply Current: 35 mA typical current draw; 100 mA available supply recommended Cable Connection: 8-pin mini DIN Wetted Parts: 303 & 302 Stainless Steel, FKM Heat Cured Silicone Rubber, Glass Reinforced Polyphenylene Sulfide, Heat Cured Epoxy, Aluminum, Gold, Silicon, Glass. Contact Omega engineering for details. -LSS (Option):316LSS, 303SS, 430FRSS, Perfluoroelastomer standard, FKM, EPDM, Buna, Neoprene Standard Pressure Drops (venting to atmosphere):
Rated 4 out of
Good Mass FlowmeterEasy to setup and reliable. Easy to use. Remote operation needs expert installation but is easy to use after setup. Price could be lower to have a broader application.
how i can connect it to o2 generator @65 psi
Asked by: asdsand
Hi, you may connect to the mass flow meter via a machine thread or NPT connection with sufficient straight run pipe or tubing, 10x the diameter of the pipe or tube upstream and 5x the diameter of the pipe or tubing downstream.