SPECIFICATIONS Accuracy: ±0.25% of span, terminal point (0.13% BFSL) Case Size: 76 mm (3'), 114 mm (4.5') Case Material: 3' stainless steel, 4.5' fiberglass-reinforced thermoplastic Case Encl. Rating: Weatherproof, NEMA 4 (IP65) Wetted Materials: 17-4 SS (sensor), 316 SS (socket) Socket Size: ¼ NPT Standard; Custom: ½ NPT, JIS, DIN, SAE, (½ NPT only with 4.5' case, others on application). Consult Engineering to order custom socket. Socket Connection: Lower (standard); 3, 9 and 12 o'clock optional Operating Temp: 10 to 60°C (14 to 140°F) Temp Error: (Zero and span) 0.08%/°C (0.04%/°F) (<0.04%°C typical) Storage Temp: -20 to 70°C (-4 to 158°F) Display: 5-digit LCD Character Height: 3' Case: 15 mm (0.60') 4.5' Case: 22 mm (0.88') Backlight: Standard Bar Graph: Standard Battery Life: 3' >1000 hr, 4.5' > 3600 hr Power Options: DC Power: 12 to 36 Vdc, 2 VA max Loop Power: 12 to 36 Vdc Electrical Rating on Optional Switch (Option S1 or S2 DPG9200 series only): (1) or (2) SPDT switches; 30 Vdc 1 A max, 125 Vac 5 A (switches adjustable to 100% of range) Backlit Display: Models 9x30 and 9x45 Battery Back-Up: Models 9130, 9230, 9145, 9245
KEYPAD FUNCTIONS: On/Off: Manually turns unit on and off (auto-off options in menu) Zero/Clear: Zeros display or clears min and max values when displayed Min/Max (Down) Arrow Key: Stores min and max values, arrow key allows for scrolling menu items Menu Key: Provides access to menu options Backlit (Up) Arrow Key: Manually turns backlite on and off (auto-off options in menu); arrow key allows for 5 menu options. (Up) arrow key allows for scrolling menu options. Enter: Selects items in the menu
MENU MODE: Engineering Units: 10 units of measurement are available: psi, inH2O (with three temp options: 20°C, 60°F, 4°C), ftH2O, mPa, mbar, kPa, kg/cm2, bar, inHg and mmHg Configuration Mode: (Config) Allows for changes to default settings of gauge, including zero disable feature Bar Graph: (Graph) Allows for adjustment of bar graph and 4 to 20 mA (optional feature) Auto Off (5 Options): (Off) allows for changes to auto-off of gauge: never, 2 min, 5 min, 15 min, 30 min Update Rate (4 Options): (Update) 100 ms, 200 ms, 500 ms, 1 s Dampening (6 Options): (Damp) None, average, 2, 4, 6, 8 times per 100 ms Backlite (5 Options): Never, 10 s, 30 s, 1 min, 5 min Field Recalibration: Allows for recalibration of zero, midscale and span (password protected)
Examples of pricing for popular models with backlit display are referenced in the 'To Order' box.
I dont remenber the passaword i have set in my DPG9145.
How can i reset this password?
Asked by: Andre12
Thank you for contacting Omega. If you have forgotten the password you entered for the DPG9145 gauge you can try using the default password which is 12045. This should allow you to enter the CONFIG menu and use the SETPW option to enter a new password. If you are still unable to access the CONFIG mode you should contact our Pressure and Strain engineers for further assistance. Thank you.