The HH147AU is a 2-channel thermometer with data logging capabilities and a capacity of 14,000 records. It offers an wide array of value added features including: accepting 7 different thermocouple types, large backlit LCD with 5-digits, dual displays (T1, T2 or T1-T2) with the maximum reading of 9999.9, relative (REL) measurement, selectable °C/°F/K units, stores 300 readings manually, max/min/avg/time and data hold functions, auto power-off with disable function, and hi/lo audible alarm functions. A USB PC serial interface and cable are also included for connectivity to a PC.
Specifications Accuracy is specified for ambient temperatures between 18 to 28°C (64 to 82°F). Thermocouple Ranges: K Type: -200 to 1372°C (-328 to 2501°F) J Type: -210 to 1200°C (-346 to 2192°F) T Type: -250 to 400°C (-418 to 752°F) E Type: -210 to 1000°C (-346 to 1832°F) R/S Type: 0 to 1767°C (32 to 3212°F) N Type: -150 to 1300°C (-238 to 2372°F) Resolution: K / J / T / E / N Type: 0.1° R/S Type: 1° Accuracy: K / J / T / E / N Type: ±(0.05% rdg, 0.7°C/1.4°F) -250 to -100°C (-418 to -148°F) ±(0.05% rdg, 0.6°C/1°F) -100 to 1372°C (-148 to 2501°F) R/S Type: ±(0.05% rdg, 2°C/4°F) 0 to 1767°C (32 to 3212°F) Temperature Coefficient: 0.01% 0.03°C of read value/°C (0.06°F/°F) outside 18 to 28°C (64 to 82°F) specified range [Under -100°C (-148°F): 0.04% of read value added to K, J, T, E, and 0.08% read value added to N] Temperature Scale: ITS-90 Battery Life: Approximately 100 hours Operation and Storage Temperature and Humidity: 0 to 50°C (32 to 122°F), < 80% RH Dimensions: 155 L x 56 W x 40 D mm (6.10 x 2.20 x 1.57') Weight: Approximately 170 g (0.37 lb) USB PC Serial Interface: Windows® XP/7/8/10 * The above specifications do not include error of thermocouple
Rated 4 out of
Nice toolWe bought it a month ago, fair enough and we're pleased with it. Perhaps it needs a better user manual but after some testings we were able to identify how some parameters should be configured.
Date published: 2017-07-17
Rated 1 out of
Does not link with our computerHad our IT guy spend an hour trying to get the unit to be seen on the computer. Never got it to work. Gave up.