HH374 meter/logger is in the end of its life cycle. As a possible substitute & upgrade, please check out Newest Released 8 Channel Thermocouple Data Logger OM-HL-EH-TC Series for more advanced features with better price.
The HH374 has a high resolution and fast analog to digital converter. Measurements settings and results are shown on the backlit LCD display. Data can be stored in the meter or directly saved on a computer through PC interface. Recorded data can be further processed on a computer. With the exclusive software, recording/recalling of data and programming of logging parameters can be done directly through a PC.
SPECIFICATIONS Measurement Range: -200 to 1370°C (-328 to 2498°F) Accuracy: -200 to 1370°C (-328 to 2498°F) ±0.1% rdg + 0.7°C (±0.1% rdg + 1.4°F) Input Protection: 60V DC or 24 Vrms AC Max Battery Type: 9V battery (included) Operating Condition: 0 to 50°C (32 to 122°F) (<80% RH) Storage Condition: -20 to 60°C (-4 to 140°F) (<70% RH) System Required: Windows NT 4.0/NT2000/XP/Vista/ Windows 7 Dimensions: 196 H x 65 W x 36 mm D (7.72 x 2.56 x 1.42') Approx Weight: 285 g (10.05 oz)
Rated 5 out of
LTD Consult from
Great portable tuning instrument!!!I purchased the HH374 about 3 months ago to as a tuning aid for my UTV/ATV DYNO and has been one of the best tuning tools I have ever purchased.
Date published: 2016-11-09
Rated 5 out of
Reliable, great battery life, stellar accuracy!reliable, great battery life, stellar accuracy!!!!
Date published: 2016-07-14
Rated 3 out of
Good productPurchased the unit several years ago and it still works great. However, I lost the software and have been requesting a new copy, or information on how to get a new copy, to no avail.
Date published: 2019-04-01
Rated 3 out of
Nice functions, but some issuesIn using the HH374 we found that the unit can be impacted by radiated noise for other systems in close proximity to the unit. Built a small shield, grounded it, and made sure not to touch in order to get stable information/data.
Date published: 2018-10-07
Rated 4 out of
Does what it ClaimsEasy to read display facilitates spot checking but battery life is less than ideal. Use of AC Adapter is essential for collection periods of more than a couple of days. Arrived with defective thermocouple which was promptly replaced by Omega.
Hi, I have a HH374 Data Logger connected via USB. When I start the real-time sampling I got an error message: INVALID FLOATING POINT OPERATION.
The Real-time set up is: 2000 data, sampling rate 10s, recording period 5:33:20 How could be fixed?
Asked by: JADA
Thank you for connecting with Omega. Please download the latest software update below. ftp://ftp.omega.com/public/TEMPERATUREGroup/products/HH374/
Answered by: Jane
Date published: 2019-03-05
Lately maybe due to updated windows 10 software, I really don't understand why, I'm not able to connect with my computer and use with the SE374 software. I've updated the software and rebooted with no luck. help!!