Please check out our newest high accyuracy Digital Pressure Gauge, the DPG210, many ranges are in stock, available for immediate delivery, and are priced competitively.
The DPG4000 takes the concept of
a test gauge to a new level. It
combines the accuracy of digital
technology with the simplicity of
an analog gauge, delivering
performance, ease of use, and a
feature set unmatched in the
pressure measurement world.
Setup of the DPG4000 is fast and straightforward, involving a menu driven
display with minimal text and
intuitive functions. The gauge is
simple enough to be used anywhere
in the world, without the need for
It also incorporates a built in data
logging function which when combined
with the DPG4000-SW software allows
it to function as a high performance
pressure data logger. The unit can
store up to 8500 data points which can
then be downloaded and stored in a
variety of formats including ASCII text,
comma delimited text, and Micorsoft
Excel Spreadsheet using the software.
The software can also be used to configure
a wide range of settings including logging
interval, number of data points per run,
capture mode and type of data.
(0 to 50ºC Unless Otherwise Noted) Accuracy: ±0.05% FS positive pressure, ±0.25% FS vacuum/500 psi and below
Temperature Compensation: 0 to 50ºC (32 to 122ºF) to rated accuracy
Standard Engineering Units: psi, bar, kg/cm2, inH2O (4ºC, 20ºC or 60ºF), ftH2O (4ºC, 20ºC or 60ºF), cmH2O (4ºC and 20ºC), mH2O (4ºC and 20ºC), kPa, mbar, inHg, mmHg, torr
Media Compatibility: Liquids and gases compatible with 316 SS
Operating Temperature: -10 to 55ºC (14 to 131ºF)
Storage Temperature: -20 to 70ºC (-4 to 158ºF)
Dimensions: 111.1 H x 127 W x 38.1 mm D (4.375 x 5 x 1.50")
Input Port: 1/4 MNPT lower mount
Display: 5 digits, 16.5 mm (0.65") high
Window: Rugged Polycarbonate
Bar Graph: 0 to 100% in 20 segments
Power: 3 "AA" alkaline batteries (included); 24 Vdc power optional
Battery Life: 1500 hr without backlight (2000 hr at slow sample rate)
Low-Battery Indicator: Icon displayed near the end of battery life. Remaining battery life can also be displayed on bar graph.
(SOFTWARE AND DATA LOGGING) Data Points: 8500 max (pressure only) Reading intervals: 1, 2, 5, 10, 30 sec, 1, 2, 5, 10, 30 min, 1 hr Data Format: Text, CSV or Excel format files Logging Modes: Continuous, Trip High, Trip Low, Trip Range Requirements: Microsoft Windows 2000 (SP4), XP (SP2) or
Vista. Exporting data to excel files requires Microsoft Excel
Version 2002 or later
† All amounts shown in USD Note: Each DPG4000 gauge comes with batteries (when applicable), protective rubber boot (with battery powered models), calibration certificate and operators manual. Lower fitting is standard. To order with rear fitting add suffix "-RM" to part number and $75 to the price.
(OPTIONS SHOWN IN THIS "NOTE" ABOVE ARE PRICED IN US$) Ordering Example: (1) DPG4000-30 Battery powered digital pressure gauge with 0 to 30 psig range, $1,195.00
Part Number Builder
Build Your Part Number Below
Range select from: 15C for -15 to +15 psig 30C for -15 to 30 psig 15 for 15 psig 30 for 30 psig 100 for 100 psig 300 for 300 psig 500 for 500psig 1K for 1000psig 2K for 2000 psig 3K for 3000 psig 5K for 5000 psig 10K for 10000 psig 15A for 15 psia 30A for 30 psia 100A for 100 psia 300A for 300 psia
Power select from: for -DC for 24 Vdc Powered Gage -RM for Rear Fitting -RM-DC for Rear Fitting with 24 Vdc Power
Note: All combinations may not be valid, check spec sheet for valid part numbers.
Rated 5 out of
Superior QualityGreat quality gauge with loads of features. The technical support was great and helped me find the perfect gauge for my testing setup. Would highly recommend this product to anyone.
Date published: 2017-07-25
Rated 3 out of
Weak link - aluminum fittingWe have 3-4 of the gauges. They are good gauges, but there is one flaw that cannot be overlooked. There is a weak aluminum fitting that connects that gauge to the process. On a couple of gauges, this fitting has cracked on us while installing, rendering the gauge useless. Recently, we had this fitting break during a 2,000 psi pressure test. We have declared this gauge unfit for use in our organization. Omega should reconsider this design.