OM-HL-SP Series include temperature only and temperature & humidity data loggers with USB interface are specifically designed for applications like Food Processing, Bio-Medical, Cold-chain transportation and Wild Life Monitoring.
The pen-sized design is IP67 rated, ensuring a high-grade waterproof and dust-proof resistance, plus the energy-saving battery can work up to 12 months before battery replacement is needed. Starting from $60, this OM-HL-SP Series offer excellent user experience to start and stop logging by Button, Delay or Specific Time. Simply plug-n-play to view, download and chart graphical data by using the free Logpro software.
Specifications Temperature Range: -30 to 70°C (-22 to 158°F) Temperature Accuracy: OM-HL-SP-T: ±1.0°C OM-HL-SP-TH: ±0.5°C Record Volume: 43,000 Points
OM-HL-SP Operation Instruction
Humidity Range: OM-HL-SP-T: NA OM-HL-SP-TH: 0-95% RH Humidity Accuracy: OM-HL-SP-T: NA OM-HL-SP-TH: ±5% RH Resolution: Temperature: 0.1°C / Humidity: 0.1% RH Power Supply: 1/2AA, 3.6V Lithium battery - included Typical Battery Life: 1 year IP Rating: NEMA 6 (IP67) Frequenty: 2 Secs to 24 Hrs Display: LCD display LCD Size: 18 L x 13.5 W mm (0.7 x 0.53") Dimensions: 125 L x 28 W x 22 Dia mm (4.92 x 1.1 x 0.86") Weight: 60 g (2.1 oz)
Logpro Recorder Analysis Software Logpro software is comprehensive temperature and humidity recorder data analysis software designed to log attribute settings, download logger data efficiently, graphically analyze data, and export the data to Excel/PDF/BMP, and other formats by needs.
The software interface is easy to use and efficient for ultimate data logging, monitoring, and analysis. Please see the Software Setup Video for more details.
If I used 32 of dataloggers can the software that comes with it handle that many. In other words can I look at a graph with all 32 sensors on it at the same time.
Asked by: FFCC
Regarding your request, you can connect as many of the units as you need to at a time, depending on the capability of your computer.
Though the loggers will only plot the data on individual graphs, they will not merge together on the same graph.
Also they are not meant to record in realtime on the computer, you set up the logger on the software, bring it to its environment, then connect back to the computer to download the data afterwards.
Answered by: JA Engineer
Date published: 2019-10-29
When the max number of readings is reached (43,000 readings), can you delete the readings and start over, or does the reader quit logging data at that point?
Asked by: Lastreaderquitat32000read
When you program the logger you can slect to either stop on memory full or start over writing the oldest data
Answered by: CW2ENGINEER
Date published: 2019-10-24
If i want to use multiple of these in different locations, anyway to streamline the time of logging between these sensors?
So i can get the same timestamp for each of these portable data loggers when reading it on the software.
Asked by: Sherlina
The logger can only do a delay or immediate start so there is no way to synchronize the time with each logger. You can look at another logger such as OM-EL-USB-2-LCD which incorporates a time delay start and than you would have merge files in EXCEL.
Answered by: PVDAS
Date published: 2019-07-26
does it come with a certificate of calibration, which is vital.
Asked by: CaTrena
Hi, regarding your question, we do have about 180 of the item in stock.
They do not include a certificate of calibration, though it can be requested upon ordering. Please email our Tech Support engineer back if you need any further assistance. Thanks!