General Purpose, Stainless Steel Pressure Transducers0 to 100 psi, Gauge, 4 to 20 mA, Cable
Please add to cart to keep it or exit the configuration
- 1, 2 & 5 psi Low Pressure Ranges!
- All Stainless Steel Construction
- Gage, Absolute, Compound Gage or Vacuum Pressure
- Rugged Solid State Design
- High Stability, Low Drift
- 0.25% Static Accuracy
- IP65 Protection Class
Edit OptionsView all models
Not all combinations are valid. Options compatible with previous selections will be in bold.
*Highlighted options are not compatible. Please select a different combination.
- Accuracy Value ±1.0% (5 psig/psia is ±1.5%
- Repeatability ±0.25% BSL
- Vibration ±20 g
- Shock 50 g
OMEGA’s PX309 Series Stainless Steel Pressure transducers below 100 psi use a high accuracy silicon sensor protected by an oil-filled stainless steel diaphragm. Units 100 psi and above use silicon strain gages molecularly bonded to the stainless steel diaphragm.
We provide a complete range of services—from product inception, through design and prototypes, to manufacturing and testing. Our application engineers work closely with our customers to customize, design or create entirely new products. Call us—whether you’re an OEM, manufacturer, or end user.
Link to Pressure Transducer Installation Article
The PX309 is also available in Metric configurations; please see the PXM309 for complete details, or contact our Pressure department for more information.
Ranges: -15 to 50 psig, 0 to 1000 psia, 100 to 10,000 psig
Accuracy (Combined Linearity, Hysteresis and Repeatability): ±0.25% BSL, max
Minimum Resistance Between Transducer Body and Any Wire: 1M Ω @ 25 Vdc
Calibration: In vertical direction with fitting down,positive outputs only
Pressure Cycles: 10 million, minimum
Pressure Overload: -15 to 50 psig and 0 to 1000 psia: 3 times rated pressure or 20 psi whichever is greater, 100 to 10,000 psig: 2 times rated pressure
Burst Pressure: -15 to 50 psig and 0 to 1000 psia: 4 times rated pressure or 25 psi whichever is greater, 100 to 10,000 psig: 5 times rated pressure
Long Term Stability (1 Year): ±0.25% of FS, typical
Operating Temperature: -40 to 85°C (-40 to 185°F)
Pressure Port: ¼-18 MNPT
Pressure Port Material: -15 to 50 psig and 0 to 1000 psia: 316 SS, 100 to 10,000 psig: 17-4 PH SS
Bandwidth: DC to 1 kHz (typical)
Shock: 50 g, 11 ms half-sine
Vibration: ±20 g
Response Time: <1 millisecond
PX309: 154 g (5.4 oz)
PX319/329/359: 100 g (3.5 oz)
IP Rating: IP65
PLEASE REFER TO 'SPECS' FOR ADDITIONAL SPECIFICATIONS FOR EACH OUTPUT (mV, 0 to 5 Vdc and 4 to 20 mA)
Are there any limitations for using the PX309 series transmitters under constant exposure to hydrogen gas (pressures up to 3k psi)? (hydrogen embrittlement specifically) Thank you.
Are FPW-15 or PSR-24 recommended as excitation sources for PX3093KG5V transmitter?
What is the input impedance and output impedance of the PX309?
What is the difference between a compound gauge and absolute gauge?
What is the accuracy and repeatability for PX359-100GI and PX319-300G5V? I see conflicting values in the product summary, common specificiations, and the datasheet.
Can this transducer be used with a 24Vdc power supply?
Is there a manual for installing and programming the PX309-5KG5V? I tried accessing the ftp.omega.com site, but it wasn't working. There also isn't a user manual on the product site. I am trying to connect it to a Arduino Uno and a 12V battery.
Can the 50 PSIa model read water pressure? or air only?