The FPD3000 Series positive displacement flowmeters are affordable and accurate. One primary feature is the ability to maintain consistent accuracy despite changing viscosity conditions. The meter’s solid construction and excellent dynamic response are well suited to the measurement of fuels and oils as well as other non-abrasive lubricating fluids. Since there is no need for straight run piping upstream or downstream of the flowmeter, the FPD3000 flowmeters are simple to use and to install. The meter has good resolution and high accuracy at low flow rates.
Specifications Accuracy: ±0.5% of reading Repeatability: ±0.03% Fitting Type: NPT: Female BSP: “-BSP” option DIN: “-DIN” option JIS: “-JIS” option ANSI: “-ANSI” option Hall-Effect Sensor Power: 4.5 to 24 Vdc (7.5 mA) Reed Sensor Power: 30 Vdc (500 mA) Output Options: Pulse Output: Standard 4 to 20 mA Transmitter: “-D-A” option; no output on battery powered “-D” model FPD3003 FPD3005D-A Both models shown smaller than actual sizer. Display: 7-digit/12 mm (0.47") upper, 7-digit/7 mm (0.28") lower all “-D” options Rate: User defined Total: Resettable Acc-Total: Non-resettable Minimum Viscosity: 1cPs Maximum Viscosity: 1000 cPs standard Maximum Pressure: See chart on page F-3 Strainer Size: See chart on next page Mounting: Shafts must be in a horizontal plane Electrical Connections: 2 x 12 mm (0.08 x 0.47"), fittings included “-D-A” option Cable Length: 1 m (3') stripped ends non-display models Mounting: Pipe Power: 4 to 20 mA, “-D-A” models 18 to 30 Vdc display, “-D” models 3 Vdc lithium battery (included) Liquid Temperature: -40 to 80°C (-40 to 176°F) Materials Body: Aluminum Enclosure: Polypropylene Seals: FKM Fasteners: Stainless steel Cable Insulation: PVC Enclosure: NEMA 6 (IP67)
COULD YOU PLEASE HELP ME WITH THE ECCN & SCHEDULE B FOR P/N: FPD3004-D S/N: D518620 FLOWMETER
Asked by: Anonymous
Thank you for your inquiry! For this request please email us at email@example.com so that we can provide this information.
Answered by: Application Engineer JM
Date published: 2021-06-30
My question is do you have calibration curve por different types of liquid or different viscosity? If my calibration certificate was made viscosity liquid 6 cPs, can I used it ](the data) with viscosity liquid 1,2 cPs (Error and incertainty)?THANKS
Asked by: Jeny
Thank you for your inquiry. Please e-mail our flow department at firstname.lastname@example.org and they will be able to assist you with providing you with additional information.