OMEGA’s LVCN210 Series ultrasonic level transmitter and controller provides continuous level measurement up to 3 m (9.8') with a 4 to 20 mA signal output, and is configured via our free software download. The level sensor has 4 programmable relays with selectable hysteresis and fail-safe logic. The embedded level controller can lower cost by replacing external control hardware. This non-contact liquid level sensor is ideally suited for corrosive, sticky or dirty liquids, and is widely selected for day tank, skid, IBC, sump and process tank level applications.
Specifications Range: 3 m (9.8') Accuracy: ±0.2% of range Resolution: 1 mm (0.039") Dead Band: 10 cm (4") Beam Width: 5.08 cm (2") Configuration: Free PC Windows® USB 2.0.The level sensor is configurable via our free PC software and USB adaptor. The sensors are offered with and without USB connectors. Connectors can be used to configure many compatible level products. Download your free copy at: omega.com/ftp Memory: Non-volatile Supply Voltage: 24 Vdc (loop) Consumption: 0.5 W Loop Resistance: 400? max Signal Output: 4 to 20 mA, two-wire Signal Invert: 4 to 20 mA or 20 to 4 mA Signal Fail-Safe: 4 mA, 20 mA, 21 mA, 22 mA or hold last Contact Type: (4) SPST relays Contact Rating: 60 VA, 1A max Contact Fail-Safe: Power loss; Hold last Echo Loss: Open, close or hold last Hysteresis: Selectable Process Temp: -7 to 60°C (20 to 140°F) Temp Comp: Automatic Ambient Temp: -35 to 60°C (-31 to 140°F) Pressure: Maximum Working Pressure = 2 bar (30 psi) Enclosure Rating: Type 6P, encapsulated, corrosion resistant and submersible Enclosure Material: Polycarbonate Trans. Material: PVDF Cable Jacket Mat: Polyurethane Cable Type: 9-conductor, shielded Cable Length: 1.2 m (48") Process Mount: 1 NPT (1" G optional) Mount Gasket: FKM Classification: General purpose Compliance: CE, RoHS
Rated 4 out of
UsefulI've used this in my college as a proof of concept. Love the easy to use system and plug and play setup.
Date published: 2017-04-10
Rated 1 out of
Not happy with this productWe have 16 of this installed on 16 genset tanks, and we can get them to give an accurate reading when the units are not running, and we had to buy voltage regulators to stabilize the spikes we get from voltage reading.
Date published: 2019-04-02
Rated 2 out of
Quit WorkingWe bought this item, installed it and it quit working the next day.